GE—Various Anesthesia Machines: May Produce Excessive Tidal Volumes under Certain Conditions
October 16, 2013 | Strategic Insights for Health System
ECRI Institute’s complete weekly summary of medical device hazard and recall information is available in ECRI Institute’s Health Devices Alerts(HDA). For more information about HDA, contact us at firstname.lastname@example.org.
In an October 4, 2013, Urgent Medical Device Correction letter submitted by ECRI Institute member hospitals, GE states that when the anesthesia units below are operating in pressure control ventilation guarantee (PCV-VG) mode, the units may produce an over delivery tidal volume when given a unique sequence of user inputs and a collapsed bellows (partially filled circuit). Exposure to excessive tidal volumes while pressure limits are maintained can result in pulmonary volutrauma and/or an increased pulmonary vascular resistance that can negatively affect cardiac filling pressure and/or stroke volume. The manufacturer has not confirmed the...