Zimmer—Air Dermatomes and Air Dermatomes II: Planetary Gear Teeth May Be Broken
May 29, 2013 | Strategic Insights for Health System
ECRI Institute’s complete weekly summary of medical device hazard and recall information is available in ECRI Institute’s Health Devices Alerts(HDA). For more information about HDA, contact us at firstname.lastname@example.org.
In a May 7, 2013, Urgent Device Removal letter submitted by an ECRI Institute member hospital, Zimmer states that the planetary gear teeth of the above dermatomes may be broken, resulting in operation failure or intermittent operation. This problem may cause a delay or postponement in surgery or result in suboptimal graft retrieval, potentially requiring an additional donor site. The manufacturer has not confirmed the...