Mindray—T1 Transport Monitor/Module Battery Packs: May Overheat
May 17, 2017 | Strategic Insights for Health System
ECRI Institute's complete weekly summary of medical device hazard and recall information is available in ECRI Institute's Health Devices Alerts(HDA). For more information about HDA, contact us at email@example.com.
In an April 21, 2017, Medical Device Correction letter submitted by an ECRI Institute member hospital, Mindray states that it has received 2 reports (from outside the U.S.) of the above battery packs overheating and forming a small, melted cavity in the outer housing. Mindray also states that in both reported instances, the battery emitted smoke and was hot to the touch, and that in 1 of the reported instances a small flame was noted on the battery surface. Mindray further states that it has received no reports of patient or caregiver injury as a result of this problem. Mindray states that if the battery overheats and results in the above monitor/modules being taken out of service, a risk to the patient may result from potential delay associated with locating another monitor. The manufacturer has not confirmed the...