Prism—C300/P300 Series Ceiling Lifts: Drive Gear Shaft May Fail and/or Free Fall Protection System May Not Engage Property
February 10, 2016 | Strategic Insights for Health System
ECRI Institute's complete weekly summary of medical device hazard and recall information is available in ECRI Institute's Health Devices Alerts(HDA). For more information about HDA, contact us at firstname.lastname@example.org.
In a September 8, 2015, Field Safety Notice letter submitted by an ECRI Institute member hospital, Prism states it has received 17 reports of incidents in which the drive gear shaft in the above lifts broke, eliminating the lift's ability to raise or lower a patient. In the 17 incidents, 3 included reports of the free fall protection system not engaging properly. If both problems occur simultaneously, the lift may unexpectedly drop the occupant. Prism also states that it has receive no reports of serious injuries associated with this problem. The manufacturer has not confirmed the...