LifeScan—OneTouch Verio IQ Blood Glucose Meters: May Turn Off during Use If User Accesses Results Log
March 14, 2012 | Risk Management News
ECRI Institute’s complete weekly summary of medical device hazard and recall information is available in ECRI Institute’s Health Devices Alerts(HDA). For more information about HDA, contact us at firstname.lastname@example.org.
In a February 21, 2012, Urgent Medical Device Recall letter, LifeScan states that the blood glucose meters below may turn off automatically when a user attempts to access the results log to view stored test results. If this problem occurs, the meter will enter set up mode when it is turned back on and require the user to confirm the date and time settings before being able to test again. LifeScan states that the issue has no effect on the accuracy of the test results or functions that include onscreen pattern notifications, averages, result tagging, or downloading, and no data is lost. However, treatment may be delayed or a treatment decision may be made in the...