Philips—Mobile Detector Holders: Loose Screws May Cause Detector to Fall
September 13, 2017 | Strategic Insights for Health System
ECRI Institute's complete weekly summary of medical device hazard and recall information is available in ECRI Institute's Health Devices Alerts(HDA). For more information about HDA, contact us at email@example.com.
In a July 14, 2017, Urgent Field Safety Notice letter submitted by an ECRI Institute member hospital, Philips states that the four countersunk screws that are used to attach the lock mechanism on the above holders may become loose, potentially causing the detector to fall down when the holder is rotated 90˚ clockwise from landscape, additional force is applied, and the moveable clamp is below the detector. The manufacturer has not confirmed the...