Philips—Xper Flex Cardio Physiomonitoring Systems: Electromagnetic Interference May Cause Electrocardiograph to Flatline
September 12, 2012 | Risk Management News
ECRI Institute’s complete weekly summary of medical device hazard and recall information is available in ECRI Institute’s Health Devices Alerts(HDA). For more information about HDA, contact us at firstname.lastname@example.org.
In a July 24, 2012, Customer Information letter submitted by an ECRI Institute member hospital, Philips states that the electrocardiograph (ECG) signal in the systems below may display a flatline when electromagnetic interference is present. This can occur when electrocautery equipment is used or when or electrophysiology magnetic mapping devices are used. The interference may result in inappropriate treatment, delay of treatment, or no treatment when treatment is indicated. Philips states that when the interferring equipment is no longer used, the ECG traces return to normal. The manufacturer has not confirmed the...