Zimmer Biomet—Orthopedic Salvage System Modular Arthrodesis Nail 0° Collar Assemblies with Locking Bolts: Locking Collar Screws May Have Dry Blast Irregularity
April 25, 2018 | Risk Management News
ECRI Institute's complete weekly summary of medical device hazard and recall information is available in ECRI Institute's Health Device Alerts. For more information contact us at firstname.lastname@example.org.
In an April 3, 2018, Urgent Medical Device Recall letter submitted by an ECRI Institute member hospital, Zimmer Biomet states that the locking collar screws packaged with the locking collar of the above systems may have a dry blast irregularity, potentially leading to surgical delay greater than 30 minutes, pain caused by implant fracture, or loosening or screw backing out, necessitating surgical intervention. Zimmer Biomet also states that this problem can be...