Terumo—6-Inch Roller Pumps Used with Advanced Perfusion System 1: May Jam during Use
March 28, 2012 | Strategic Insights for Health System
ECRI Institute’s complete weekly summary of medical device hazard and recall information is available in ECRI Institute’s Health Devices Alerts(HDA). For more information about HDA, contact us at email@example.com.
In a March 16, 2012, Urgent Medical Device Recall letter submitted by an ECRI Institute member hospital, Terumo states that it has received reports of instances in which the roller pumps below displayed a "pump jam" message and stopped while in use. Terumo states that pump jam is more likely to occur when using the 6-inch roller pump than when using the 4-inch roller pump for dual-sized tube set cardioplegia delivery. In addition, the occlusion settings suggested in the Instructions for Use (IFU) for some...