Cardinal Health—Halyard Health KimVent Closed Suction Systems for Adults: May Contain Small Hole, Potentially Allowing Air to Leak out of Closed Ventilator Circuit and Triggering Ventilator Alarm
January 28, 2015 | Risk Management News
ECRI Institute's complete weekly summary of medical device hazard and recall information is available in ECRI Institute's Health Devices Alerts(HDA). For more information about HDA, contact us at firstname.lastname@example.org.
In a January 21, 2015, Urgent Product Recall letter, Cardinal Health states that the suction systems below may contain a small hole, which may be difficult to visually identify, on the double-swivel elbow. If the hole allows air to leak out of the closed ventilator circuit, the ventilator may be unable to deliver the programmed air pressures and the ventilator alarm may be triggered. The manufacturer and distributor have not confirmed...