Evaluation: dBMEDx BBS Revolution Bladder Scanner
December 21, 2018 | Evaluations & Guidance
Here's how the dBMEDx BBS Revolution compares with the other bladder scanners we've evaluated.
Click the product names below to view our complete findings. Products are listed alphabetically by supplier.
The dBMEDx BBS Revolution is a good choice for bladder volume applications. The BBS Revolution does not have a B-mode ultrasound display—a feature typically used on bladder scanners to guide probe placement for volume data acquisition. Instead, this scanner has a wide probe and 180° scan angle to acquire volume data over a large anatomical region. The scanner met our preferred criteria for accuracy in tests using commercially available test phantoms. The BBS Revolution's wireless probe offers advantages in terms of being easy to clean and use. The scanner also tested well in the categories of workflow, interoperability, cybersecurity, and maintenance.The BBS Revolution. (Image courtesy of dBMEDx.)
- The purpose of the BBS Revolution is to calculate an estimate of urinary bladder volume. There are no supplementary...